June 4, 2018¶
Status of Issues¶
- Channels with holes in the ADC distribution (VP)
- How many channels?
- Run range
- Any other bad channels/features?
- Need code for blacklisting
- Implement code in calibration (VP)
- Implement code in reconstruction (DR)
- Status (VP/SW)
- Status of improved tw fits (SW/VP)
- VP reported there were some issues with weird TW distributions, and issues getting the T0 calibration to work.
- VP to check with SW on data used for calibration input
- Need to verify/understand:
- is the dT offset issue resolved?
- what is the TOF2 efficiency
- 0.5ns cut should be loosened
- New cut = ? (VP)
- Implement new cut (DR)
- Mon June 18, 9am